Catheter material consists of sterile 3.5 Fr polyurethane (PU) tubing with a 2 Fr PU tip. The catheter is sealed with a sterile stainless steel pin. 22 gauge blunted needles are required to access the port. Fill volume of the catheter is 90µl.
Heparinized Glycerol (500 IU/ml): 10.0 mL stock heparin (1000 IU/mL) + 10.0 mL 99% Glycerol solution (Sigma).
To maintain animals over longer periods of time, catheters need to be flushed twice a week (once every 3-4 days). Flush catheters by following the sampling procedure below, minus the withdrawal of the whole blood sample.
Blood sample size can vary, but recommendations are in the range of 100-200µl. Sample size and frequency should be minimized to essential time points to maintain the health of the animal. For blood withdrawal, gather the following materials: Syringe assemblies (1cc syringe attached to a 22G blunted needle), sterile saline and sterile fill solution.
Place animal in a restrainer (small open topped boxes the size of a pipette container work well).
Important: Always clamp the port with rubberized or smooth hemostats to prevent unintended blood flow and port damage when changing syringes and flushing the catheter.
Clamp port and remove the pin from the catheter and set aside. Insert an empty syringe assembly (SA) into the port and release hemostats.
Gently withdraw fill solution and blood; clamp port
Attach a second SA, release clamps and withdraw sample (syringe may contain anticoagulant); clamp port.
Release hemostats and slowly flush catheter with sterile saline ~200µl (or greater to match blood withdrawal); clamp port.
Add 90µl of sterile lock solution, clamp port and replace pin.
If blood fails to flow in step 4, remove the empty SA and replace with a SA containing sterile saline and gently flush the catheter. Continue as outlined above.
Individually house animals to prevent cage mates from chewing on one another’s catheters.
Staples should be removed 7-10 days post-operatively; do not remove staple around the catheter port.
- During animal manipulation (dosing / weighing), it is important not to place undue stress on the catheter.
- Using needles larger than 22 gauge will stretch the port and make sampling difficult. Additionally, sampling by means of needles with bevels or rough edges will damage the port, again making sampling difficult.